November 28, 2016
Thank you for using CloudTesting™ Service.
There is a problem on current CloudTesting™ Lab. Uninteded power-on may be performed.
- Pattern ViewerTool (PCXT01-M6010)
- Logic Analyzer Tool (PCXT01-M6030)
When these analysis IPs are started, device powers are forcibly turned on. In this case, PBVCC is not be turned on.
If you replace the device or operate the measurement board immediately after starting these analysis IPs, the device may be damaged.
The improved version will be released as soon as possible.
Until the release, when starting these analysis IPs, take the following. The power shutdown sequence is executed by this operation.
- Execute a flow